A short review about new era NE-1002X microfluidic syringe pump
introduction about new era NE-1002X
Outside of conventional syringe pump which generate flow oscillation at low flow rate and long settling time we can now found syringe pump dedicated to microfluidic experiment. In theory a microfluidic syringe pump do not generate visible flow oscillation even at low flow rate. To learn more about syringe pump and microfluidic and how to optimize their eventual side effects read our tutorial about the use of syringe pump in microfluidics. The NE- 1002X syringe pump from New Era is , to our knowledge, the cheaper microfluidic syringe pump for microfluidic. ( For instance Darwin Microfluidics offers syringe pumps ERA for microfluidics )
Need advice to choose your microfluidic flow control?
Feel free to contact us at:
Ask me your question
(We will answer within 24 hours)
Flow oscillation and stability
Lets come back to flow oscillation, contrary to the Cellix Exigo, New Era do not provide in their datasheet the minimum pulseless flow rate of their pump.
We can found in their datasheet the minimum flow rate they can reach depending of the syringe diameter and the New Era 1002X can reach flow rate down to 14nL/hr with a 1mL BD syringe to 0,008 nL/hr with an 1µL Hamilton glass Microliter syringe.
The New Era 1002X syringe pump use step motor with 400step per lead screw revolution which is clearly lower than the Cellix one ( 250 000 steps ), The new Era also use a 25:1 gear box. The minimum advance per step is comprised between 4,2 nm and 34 nm depending on motor speed.
Response time and settling time
Contrary to Cellix Exigo syringe pump the New Era syringe pump does not include a flow rate feedback loop to fasten flow switch in your microfluidic device. If you want more information about how to reduce flow oscillation and slow responsiveness of your syringe pump discover our tutorial about the use of syringe pump in microfluidic.
Additional information about New Era NE-1002-X syringe pump
Front Panel control
Contrary to Cellix Exigo syringe pump you can control the New Era syringe pump using front panel control.
Contrary to Cellix Exigo syringe pump, The New Era syringe pump do not provide software but only a terminal emulator and programming libraries. The new Era syringe pump provides a RS-232 port for programming.
The pump include internal and external trigger for synchronization with other instruments
If you want more information about the New Era NE 1002 X syringe pump, you can contact Barry Cowen from syringepump.com or contact us.
Discover other microfluidic syringe pump of our review : Cellix ExiGo syringe pump, Harvard Apparatus Phd ultra advanced syringe pump, Harvard Apparatus 11 pico plus Elite syringe pump, KD Scientific Legato 180 Picoliter Syringe Pump.
WORLD LEADER IN HIGH PERFORMANCE MICROFLUIDIC FLOW CONTROL
We provide the only microfluidic flow control system using Piezo technology that enables a blazing fast flow change in your microdevice.